Pharmaceutical Stability Test Chambers Description and Specifications for Lab Tenders
Pharmaceutical Stability Test Chambers
The cabinets should meet the requirements of ICH Guidelines Q1A (Stability Testing) and Q1B (Photostability Testing).
Temperature range: +10 to +60°C
Humidity range: 20 to 90%RH
Temperature and Humidity accuracy ± 0.3°C / ±2.5%RH.
Temperature and Humidity uniformity ± 0.5°C / ±3.0%RH.
PID Cotrolled with LCD display.
Built in Digital Recorder.
Balanced Temperature and Humidity Control system (B.T.H.C).
Running Hours of operation counter.
Safety Device: ELB, Refrigerator Overload Relay, Overheat Protector, Water Empty